Air Conditioner Test unit (AHRI210/240)
Synopsis of the EquipmentThis equipment is for measuring the calories of Air-con. It also measures various kinds of data including pressure and electric properties of the Air-con. And it consists of indoor laboratory (1 room), outdoor laboratory (1 room), air handling unit, refrigerating machine, airflow meter, power panel, computer system and so on.
Definition of the Test Unit
Window type Air-con.
Split type Air-con.
Cassette type Air-con.
Floor standing type Air-con.
Fan coil unit
Cooling capacity 9, 000BTU/h ~60, 000BTU/h (2, 250kcal/h~15, 000kcal/h).
Heating capacity 9, 000BTU/h ~60, 000BTU/h (2, 250kcal/h~15, 000kcal/h). Amount of dehumidify max.
15 L/h (indoor)
Water flow 0~4m3/h
Air flower rate (Indoor) 5~60m3/min(175~2, 100CFM)
Power 1 60~300V 40A continuous rating (at 230V).
3 60~600V 30A continuous rating (at 380V).
Size Indoor unit no bigger than W×D×H=1200×800×1900MM
Outdoor unit no bigger than W×D×H=1200×800×1200MM
Window unit no bigger than W×D×H=800×800×800MM
Inspection ConditionsRepeatability Test (by heater)
Test method ------will test 2 times at 3 grade.
Acceptable heater difference ----- 1.5% beneath.
Standard sample as compared with to test (by ETL unit).
Test method ------will test 3 times at 3 grade of ETL sample.
Acceptable heater difference ------±1.5% compared to ETL Test result.
For fan coil unit the balance between water side and air side ------5% beneath
Conditions other than these will match the temperatures established in the correlative standards and rules